I have, for quite a while, provided modified versions of the code for my projects free of charge. However, the cost of webhosting for this site has tripled in the last couple of years and the little I make out of the pcbs I sell goes nowhere near covering this cost.
As of October 2017, I will be asking for a donation in exchange for providing custom modified code for my projects to help keep this site available.
As a lot of people have pics of things unrelated to ham radio on their websites, I thought I would add a couple of my own (Scroll down for site updates). Banjo, our Red Heeler/Koolie cross at 4 months old. And at 12 months old (Feb 2016) enjoying his birthday breakfast treat of banana on toast. A new addition to the family, an 11 week old kitten (end July 2017), rescued by Banjo after being trapped between a water tank and fence. Her name is Storm after her colouring and it's eating a bit of toast and jam (yes, our animals do eat strange things, including lemon meringue pie). They are now the best of friends.
Simple DDS VFO project updated and software file uploaded. The pushbutton on the encoder is used to change step size (step sizes 10Hz, 1kHz and 10kHz).
Noise Canceller kits now available.
Updated both the DDS Xtal Substitute and DDS Xtal Substitute Mark 2 projects into one project. The original webpages are available to download as PDF's.
New project added - my version of the X-Phase noise canceller. PCB available and a kit of parts to follow soon
27 December 2016
Another addition to the Gallery section. This time, fitting a DDS module to a Philips PM5326 RF Generator to cure a bad drift problem.
This is a mod using a DDS module to repair a drifty Philips PM 5326 RF generator. Stefan has done a marvellous job, not only of the mod, but also in providing the write-up and pictures presented below.
Starting point for this project was a constant frequency drift of my Philips PM 5326 RF generator. Even after hours of warming up it doesn't come to a full stop. Furthermore tuning to a specific frequency was a touchy thing because no fine tuning was possible with the build in tuning capacitor. The RF section of the generator is based on the PLL principle. The reference frequency is generated by a VFO, comprising a simple transistor LC oscillator. This reference frequency determines the frequency of a VCO as the active element of the generator. It generates the RF frequency between 100 KHz and 125 MHz via several range coils and divider chains for the phase comparator. Every slight deviation in frequency of the VFO results in a relatively higher deviation of the VCO and therefore of the RF frequency. So, the weak point was the VFO, which was state of the art in the late 70's, early 80's. It was as good as was possible in those days. So, I decided to replace the "analog" VFO with a modern DDS-VFO. An additional requirement was, that the generator shoud remain in its original state. I never wanted to drill any holes to the case for adding any switches etc. The modification should not be seen from outside and it should be reversible. Browsing through the internet I found the "Simple DDS VFO" from Terry (VK5TM) which fits all my needs, except for the step size switch. But Terry thankfully updated the software for the microcontroller, so that the step size could be toggled with a push switch on a suitable encoder. The following pictures show how the project was realised: Pic_00: The circuit diagram, modified for my project. Power supply was altered and 100n cap fitted across the step switch. Pic_01 + Pic_02: The DDS-VFO circuit build on a Veroboard. Pic_03 + Pic_04: Aluminium die-cast case, ready to mount. Pic_05 + Pic_06: A look into the Philips PM5326 RF Generator. You can see the RF-section opened with the VCO (0,1-125 MHz) and the VFO (498-1255 KHz) parts. On the right side the tuning capacitor for the VFO. It has to be removed. Pic_07: The tuning capacitor removed. A lot of work to do that. The side panel has to be removed as well as the front plate of the generator. Pic_08: The empty space where the case for the DDS-VFO will be installed. Pic_09 - Pic_11: The rotary encoder all ready in place with a new bracket. Pic_12 + Pic_13: The generator's front and side panel taken apart. Pic_14 + Pic_15: The new die-cast case in place and the side panel mounted. Pic_16: All the knobs fitted on the front plate. Pic_17: The circuit built into the aluminium case. Pic_18 + Pic_19 : The microcontroller 12F629 and the new wiring harness installed. Pic_20: Finally the AD9850 DDS-module on top of the veroboard. Pic_21 + Pic_22: Alignment of the DDS-VFO to the center frequency 876 KHz with my Racal Dana 1991 frequency counter (with OCXO option). Frequency is spot on after 30 min. warm up time). Pic_23: Fit the cover and tighten the screws of the DDS-VFO case. Pic_24: The same for the RF section of the generator. Pic_25: Wire harness of the encoder instead of the tuning capacitor's shaft. Pic_26: The generator completely assembled and in its case. No optical changes visible. But the frequency setting knob now has a push button (built within the encoder). Pic_27 + Pic_28: The generator turned on and tuned to the FM and AM IF frequencies 10,7 MHz and 455 KHz repectively. Frequencies now are absolutely stable and precise (tolerances of only a few hertz). A huge improvement to the former state ! The modification was worth all the efforts. I am very grateful to Terry for all his support and help making this project a great success.